This product is composed of the glass fiber strips of monoclonal antibody against FSH.Anti-mouse IgG solid cellulose nitrate memberance and the bonders of absorptive colloidal gold-monoclonal antibody are against FSH. It adopts the principle of double antibody sandwich method and immunochromatography to test the FSH in the urine.
1.To begin testing; open the sealed pouch by tearing along the notch..Remove the test from the pouch.Immerse the strip into the urine with the arrow end pointing towards the urine. 2.Do not immerse past the MAX (maximum) line. Take the strip out after 3 seconds and lay the strip flat on a clean, dry, nonabsorbent surface (e.g., mouth of the urine container 3.Wait for colored bands to appear. Depending on the concentration of FSH in the test specimen,positive results may be observed in as short as 40 seconds. However to confirm negative results,the complete reaction time (5 minutes) is required. Do not read the results after 10 minutes. 4.Re-cap the device and wait for color band to appear. Read results within 5 minutes
Interpretation of Results:
POSITIVE: Two lines are visible and the test line (T) is the same as or darker than the control line (C). A positive result indicates that FSH levels are higher than normal and the subject you may be experiencing peri-menopause. NEGATIVE: Two lines are visible, but the test line (T) is lighter than the control line (C), or there is no test line.A negative result indicates that the subject is probably not experiencing peri-menopause in this cycle. INVALID: Control line (C) fails to appear. Insufficient specimen volume or incorrect test performance are the most likely reasons for an invalid result. Review the procedure and repeat with a new test. If the problem persists,discontinue using the test kit immediately and contact
1. Format: strip,cassette,midstream
2. Accuracy: >99%
3. Reaction Time:5 minutes
4. Fast to test
5. Easy to operate
6. Sensitive and accurate result
7. Competitive price
Q1: What's the method u use to do the test? A: Colloidal Gold Method.
Q2: What's ur MOQ ? A: Always, for strip format, the MOQ is 5000pcs. While for cassette format, MOQ is 2000pcs, same as midstream.
Q3: What's ur packing of the tests? A: If u choose the low cost, we will suggest u the bulk packing to go. Which means 1 test in a single pouch, then 100/40 pouches in a plastic bag, 50 bags in a carton. If u need to sell in pharmacy/chain shop or some countries which have strict requirements about the packing, u can choose to pack in a box, like 100pcs in a box, or even 1 test in a single box.
Q4: What's the main differences when comparing products from different suppliers? A: U can compare the mainly features of the test: Accuracy, Sensitivity, Specificity, and then the Price.
Q5: Some tests can be tested with serum/plasma or with whole blood, so what's the difference? A: Well, if the specimen is whole blood, people will need buffer, lancet and alcohol swab to help the test. If they buy all the accessories, they can test very easily at home. But if the specimen is serum/plasma, it will need centrifuge to separate serum from whole blood first. This way, it's better to use at lab or hospital. And in some words, serum/plasma test will always give a more accurate results.
Q6: How can i distinguish a good test kit? A: U can judge from the 4facts:Technical data: Such as the accuracy, specificity and sensitivity.
Pouch sealing: Tight enough. If the foil pouch is not sealed well, the humidity in circustance will destroy the reactivity of antibodies labeled on NC membrane. Shelf life will be shorten down. Flow speed: The shorter the better?? NO!! The reaction of antibodies on NC membrane and antigens in specimen usually requires quite a while to work sufficiently. U can refer to the instruction for more accurate time. Background: Good test usually gives clean background after running. If there are red smears in the reading window, it usually caused by bad colloidal gold technology or bad NC membrane. Sometimes, the defect caused false positive result in practice